# MPP300 Rel /03/11

Maat: px
Weergave met pagina beginnen:

## Transcriptie

1 Accessory for checks on PV plants with multi MPPT Pag 1 of 4 1. NOTES ON MPPT (Maximum Power Point Tracker) Solar irradiation on a surface such as the surface of a photovoltaic system has extremely variable characteristics, since it depends on the position of the sun with respect to the surface and on atmospheric conditions (typically, on the presence of clouds). A photovoltaic module presents, for different solar irradiation values, and for different temperature values, a range of characteristic curves of the type shown in the following figure. In particular, the figure shows three I-V curves (in bold) which correspond to three values (1000, 800, 600W/m 2 ) of solar irradiation On each characteristic curve there is one single point in which the power transfer towards a hypothetical charge supplied by the photovoltaic module is maximized. The maximum power point corresponds to the voltage-current pair for which the product V*I is maximum, where V is the value of voltage at the module s terminals and I is the current which runs in the circuit obtained by closing the module on a hypothetical charge. With reference to the figure above, the product V*I is represented, for the three solar irradiation values mentioned above, through the three curves in thinner lines. The figure shows that, as stated above, these curves only have one single maximum point. For example, for 1000W/m 2, the maximum power point corresponds to a voltage value of approx. 36V and to a current value of approx. 5.5A. Obviously, if the power provided by the system is maximized, it is possible to make the most of the system, both in case the system is connected to mains, and in case it is stand-alone. MPPT is an inbuilt device in the inverters. It typically reads the voltage and current values at any instant, calculates their product (i.e. the power in Watts) and, by causing small variations in the conversion parameters (duty cycle), it is capable of determining, by comparison, if the photovoltaic module is working in maximum power conditions or not. According to the result, it operates again on the circuit in order to bring the system to an optimal condition. The reason why MPPTs are used is simple: a photovoltaic system without MPPTs may operate anyway. However, with the same solar irradiation, it provides less energy. There are inverters with 1, 2 or also 3 inbuilt MPPTs available on the market. Typically, the inverters with more than one MPPT are used in systems where: The different photovoltaic systems it consists of forcibly have different inclinations or directions. In this way, each single MPPT manages its own photovoltaic field, maximizing its performance for the corresponding irradiation and temperature characteristics (without being influenced by the other photovoltaic fields) A greater service continuity is sought. With more MPPTs it is possible to put one single photovoltaic field out of service, while the others continue producing energy towards the remaining MPPTs.

2 Accessory for checks on PV plants with multi MPPT Pag 2 of 4 2. DESCRIPTION OF MPP300 ACCESSORY The MPP300 model has been designed as an exclusive accessory of a Master SOLAR300N and SOLAR I-V instruments for the purpose of carrying out testing simultaneous operations on single-phase and three-phase PV systems with up to 3 Multi MPPT device Together with a Master instrument (see enclosed principle scheme) MPP300 is the ideal solution for testing and analyzing the possible problems linked to possible low efficiency values of photovoltaic systems The instrument has the following features: Testing of PV systems with single/multi-mppt inverter - single/three-phase AC output Using with Master instruments SOLAR300N and SOLAR I-V Measurement of 3 DC voltages and currents Measurement of DC string power and total DC power Measurement of 3 AC TRMS voltages and currents Measurement of total AC power Measurement of irradiation [W/m 2 ] by means of a reference cell connected to unit SOLAR-02 Measurement of panel and environmental temperature by means of probe PT300N connected to SOLAR-02 Parameter recording of a PV system with 5s to 60min programmable IP Operations with LED indication Internal memory for data saving RF interface for transferring the data to the SOLAR I-V instrument and SOLAR-02 remote unit USB interface for transferring the data to the SOLAR300N instrument

3 Accessory for checks on PV plants with multi MPPT Pag 3 of 4 3. TECHNICAL SPECIFICATIONS (*) Accuracy is indicated as [%reading + (no. of digits) * resolution] at 23 C ± 5 C, <80%HR DC Voltage Range (V) Resolution (V) Accuracy (0.5 rdg + 2dgt) AC TRMS Voltage Phase-Neutral Single/Three phase systems Range (V) Frequency (Hz) Resolution (V) Accuracy (0.5 rdg + 2cifre) Max. crest factor: 1.5 AC TRMS Voltage Phase-Phase Range (V) Frequency (Hz) Resolution (V) Accuracy (0.7 rdg + 2dgt) Max. crest factor: 1.5 DC Current by means of external clamp transducer STD type Range (mv) Resolution (mv) Accuracy 5.0mV 319.9mV (0.5 rdg %FS) 0.1mV 10V 320.0mV 999.9mV 0.5 rdg Current values corresponding to a voltage < 5mV are zeroed AC TRMS Current by means of external clamp transducer STD type 5.0mV 219.9mV (0.5 rdg %FS) mV 220.0mV 999.9mV 0.5 rdg Max. crest factor: 1.5 ; Current values corresponding to a voltage < 5mV are zeroed 10V AC TRMS current (by means of external clamp transducer of FLEX 8.5µV/A FS 100A) mV Hz 0.001mV (0.5%rdg + 7dgt) 10V Max. crest factor: 1.5 ; Current values < 1A are zeroed AC TRMS current (by means of external clamp transducer of FLEX 8.5µV/A FS 1000A) mV Hz 0.01mV (0.5%rdg + 15dgt) 10V Max. crest factor: 1.5 ; Current values < 5A are zeroed DC Power (Vmis > 150V) Parameter FS clamp (A) Range [W] Resolution [W] Accuracy 0.000k 9.999k 0.001k 1< FS k 99.99k 0,01k (0.7 rdg+3dgt) DC POWER 10< FS k 100.0k 999.9k 0.0k 999.9k 100< FS k 9999k Vmis = voltage at which power is measured; Imis = measured current 0.01k 1k (Imis < 10%FS) (0.7 rdg) (Imis 10%FS)

4 Accessory for checks on PV plants with multi MPPT Pag 4 of 4 AC Power (Vmis > 200V, PF=1) Parameter FS clamp (A) Range [W] Resolution [W] Accuracy 0.000k 9.999k 0.001k 1< FS k 99.99k 0,01k (0.7 rdg+3dgt) AC POWER 10< FS k 100.0k 999.9k 0.0k 999.9k 200< FS k 9999k Vmis = voltage at which power is measured; Imis = measured current 4. GENERAL SPECIFICATIONS 0.01k 1k (Imis < 10%FS) (0.7 rdg) (Imis 10%FS) POWER SUPPLY: Internal power supply: Battery duration: External power supply: Internal rechargeable Li-ION battery (3.7V, 1900mAh) > 3 hours AC/DC adapter V 50/60Hz / 5VDC CHARACTERISTICS OF RADIO MODULE Frequency range: GHz R&TTE category: Class 1 Max transmission power: 30µW Max RF connection distance: 1m MEMORY AND EXTERNAL INTERFACE Memory capacity: 2Mbytes Integration Period (IP): 5,10,30,60,120,300,600,900,1800,3600s RF interface: connection SOLAR I-V and SOLAR-02 units USB interface: connection SOLAR300N unit Battery duration (with SOLAR-02): approx. 1.5 hours PI = 5s) ; approx. 8 days PI = 600s) MECHANICAL CHARACTERISTICS Dimensions (LxWxH): Weight (battery included): Protection index: 300 x 265 x 140 mm 1.2 kg IP40 ENVIRONMENTAL CONDITION FOR USE Reference temperature: 23 C 5 C Operating temperature: 0 40 C Allowable relative humidity: <80%hr Storage temperature: C Storage humidity: <80%HR REFERENCE STANDARDS Safety: IEC/EN Safety of measuring accessories: IEC / EN Insulation: double insulation Pollution level: 2 Measurement category: CAT III 1000V DC, Max 1000V between DC inputs CAT IV 300V AC to ground, Max 600V between AC inputs Max operating altitude: 2000m This instrument satisfies the requirements of Low Voltage Directive 2006/95/EC (LVD) and of EMC Directive 2004/108/EC (*) Technical specifications can be modified without preliminary advise

5 EURO-INDEX Service Onderhoud en kalibratie van meetinstrumenten... EURO-INDEX beschikt over een bijzonder modern service- en kalibratielaboratorium. Hier worden de meetinstrumenten uit het assortiment preventief onderhouden, gerepareerd, gekalibreerd en indien nodig gejusteerd. Het service- en kalibratielaboratorium van EURO-INDEX is verdeeld in verschillende disciplines, gebaseerd op het soort meetinstrument en de gemeten grootheden. Druk Gasanalyse Temperatuur (inclusief infrarood temperatuurmeting en thermografie) Elektrische grootheden Gasdetectie Luchtsnelheid en luchthoeveelheid Waarom een kalibratiecertificaat?... Een kalibratiecertificaat vermeldt hoeveel een meetinstrument afwijkt ten opzichte van onze, naar (inter)nationale standaarden herleidbare, kalibratiemiddelen. Bij de meetresultaten op het certificaat wordt tevens vermeld of het meetinstrument voldoet aan de specificaties die door de fabrikant zijn opgegeven. Zonder kalibratiecertificaat kunt u er vanuit gaan dat de meter voldoet aan de fabrieksspecificaties, maar aantonen kunt u dit niet. Een testcertificaat van de fabrikant is te beknopt om de lineairiteit aan te tonen en is niet geregistreerd op naam (wat wel degelijk een vereiste is). SPECIMENEN SPECIMEN KWS... KWS is een uniek servicesysteem voor uw meetinstrumenten met periodiek onderhoud en kalibratie. Veel zaken worden voor u geregeld, zodat u zonder zorgen gebruik kunt maken van uw meetinstrumenten. De prijs staat vast voor de levensduur van het instrument (mits de KWS behandeling volgens herkalibratieadvies periodiek wordt uitgevoerd in het EURO-INDEX kalibratielaboratorium) Geen arbeidsloon bij de KWS behandeling Kalibratie voor justage (voorkalibratie) indien mogelijk Indien nodig justage en (na)kalibratie Reparatie en preventief onderhoud Gratis oproep met het advies voor herkalibratie Controle op functionaliteit van het instrument Vijf jaar historie voor alle gegevens 10% korting op onderdelen Serienummerregistratie Franco retourlevering RvA accreditatie... Het kalibratielaboratorium van EURO-INDEX beschikt sinds 21 augustus 1997 over een RvA accreditatie naar NEN-EN-ISO/IEC Deze accreditatie geldt voor verschillende grootheden, zoals gespecificeerd in de scope bij accreditatienummer K105 op Test- en meetinstrumenten voor grootheden die deel uitmaken van de gespecificeerde scope, kunnen worden voorzien van een RvA kalibratiecertificaat. De metingen worden uitgevoerd met standaarden waarvan de herleidbaarheid naar (inter)nationale standaarden, ten overstaan van de Raad voor Accreditatie, is aangetoond. In het Multilateral Agreement zijn de meeste Europese landen overeengekomen elkaars accreditaties te accepteren. Hierdoor is een RvA kalibratiecertificaat internationaal geaccepteerd. Bovendien wordt op een RvA kalibratiecertificaat de meetonzekerheid van de gerapporteerde meetresultaten vermeld. Verhuur van meetinstrumenten... EURO-INDEX biedt een assortiment meetinstrumenten te huur aan. Na deskundig advies van onze productspecialisten, wordt bepaald welk instrument u nodig heeft voor uw specifieke werkzaamheden. De instrumenten worden compleet met accessoires geleverd, inclusief herleidbaar kalibratiecertificaat. Wijzigingen voorbehouden EURO-INDEX NL NEDERLAND Rivium 2e straat LG Capelle a/d IJssel T: F: Leuvensesteenweg 607 BELGIË 1930 Zaventem T: (0) F: (0)

### SOLAR I-V Rel. 1.01 20/07/10

PV single phase certifier and I-V curve checker Pag 1 of 5 1. SOLAR I-V MAIN FEATURES SOLAR I-V: use as PV certifier SOLAR I-V performs all tests on Single phase PV plants by using of SOLAR-02 remote unit

### PVCHECKs Rel /09/15

Meter for safety, functionality and performance on PV plants Pag 1 of 6 The multifunction instrument PVCHECKs performs prompt and safe electrical checks required for a PV system (DC section) and controls

### SOLAR I-Vw Rel. 1.03 06/10/15

PV single phase certifier and I-V curve checker up to 15A Pag 1 of 5 1. SOLAR I-Vw MAIN FEATURES SOLAR I-Vw: use as PV certifier SOLAR I-Vw performs all tests on Single phase PV plants by using of SOLAR-02

### SECUTEST BASE VEILIGHEIDS BASE10 DE NIEUWE SECUTEST VEILIG EN EFFICIËNT TESTEN VAN ELEKTRISCHE ARBEIDSMIDDELEN

DE NIEUWE SECUTEST SECUTEST BASE VEILIGHEIDS BASE10 VEILIG EN EFFICIËNT TESTEN VAN ELEKTRISCHE ARBEIDSMIDDELEN Elektrische apparaten volgens NEN3140 Incl. verlengkabels en (P) RCD s Medische apparaten

### SOLAR300N Rel. 1.03 21/01/10

Photovoltaic installation certifier + professional network analyzer Pag 1 of 5 1. SOLAR300N MAIN FEATURES SOLAR300N performs all tests on PV plants by using of SOLAR-02 remote unit which, after a preliminary

### Technical data sheet EX-TEC HS 680

Technical data sheet EX-TEC HS 680 Device data Dimensions (W x D x H) Weight approx. 148 x 57 x 205 mm approx. 148 x 57 x 253 mm with supporting bracket approx. 1000 g, depending on equipment Certificates

### SOLAR I-Ve Rel /05/17

I-V curve tracer up to 15A or 1500VDC and PV efficiency logger Pag 1 of 4 1. ELECTRICAL SPECIFICATIONS Accuracy is calculated as [ readings + (no. of digits) * resolution] at 23 C 5 C, con relative humidity

### BIMETAAL THERMOMETERS THERMOMÈTRES BIMÉTALLIQUES

STATIONAIRE THERMOMETERS DIGITALE TEMPERATUURMETERS THERMOMÈTRES STATIONNAIRES THERMOMÈTRES DIGITAUX Algemene informatie / Information générale Standaard stationaire thermometers kunnen worden voorzien

### HANDLEIDING BLAUWE LIJN EUROprinter infrarood

HANDLEIDING BLAUWE LIJN EUROprinter infrarood INHOUD 1. AAN- EN UITSCHAKELEN VAN HET APPARAAT...2 2. PRINTERPAPIER PLAATSEN...2 3. BATTERIJEN...3 4. INSTELLINGEN PRINTER...3 5. GEBRUIK IN COMBINATIE MET

### PVCHECKs Rel /10/15

Safety, commissioning and performance tester of PV plants Page 1 of 6 The multifunction instrument PVCHECKs performs prompt and safe electrical checks required for a PV system (DC section) and controls

### KAN DE HITTE AAN. EN STOF, WATER EN EEN VAL VAN 3 METER. Introductie van de nieuwe 62 Max infraroodthermometers van Fluke.

KAN DE HITTE AAN. EN STOF, WATER EN EEN VAL VAN 3 METER. Introductie van de nieuwe 62 Max infraroodthermometers van Fluke. Stof- en waterdicht: Conform beschermingsklasse IP54 en dus stof- en waterdicht.

### NEPTUNE Rel 1.02 of 17/10/18

Multifunction professional safety multimeter Page 1 of 4 Multifunction professional safety multimeter Page 2 of 4 1. ELECTRICAL SPECIFICATIONS Accuracy calculated as [%reading + (num. dgt* resolution)]

### Klein formaat, grote prestaties AMP-25-EUR AC/DC TRMS Mini-Stroomtang

Herfst/winter 2015 Periode: 01.09.2015 31.12.2015 Informatie over de nieuwste producten en acties van BehaAmprobe. Klein formaat, grote prestaties AMP25EUR AC/DC TRMS MiniStroomtang Kabels zoeken opnieuw

### STATIONAIRE MANOMETERS MANOMÈTRES STATIONNAIRES

Algemene informatie / Information générale Stationaire manometers kunnen worden voorzien van uw bedrijfslogo. Prijzen op aanvraag. KWS (kalibratie) is mogelijk op stationaire manometers tot 25 bar (code

### Database Memory HANDLEIDING

Database Memory HANDLEIDING Handleiding BLAUWE LIJN Database Memory INHOUDSOPGAVE 1. OVER DEZE HANDLEIDING... 2 2. KLANTGEGEVENS... 2 2.1. Invoeren van klantgegevens... 2 2.2. Klantgevens aanmaken/bewerken...

### HANDLEIDING BLAUWE LIJN RHT-serie. temperatuur- en luchtvochtigheidmeters

HANDLEIDING BLAUWE LIJN RHT-serie temperatuur- en luchtvochtigheidmeters INHOUD 1. KENMERKEN... 2 2. BEDIENING EN WERKWIJZE... 2 2.1. Batterij... 2 2.2. Temperatuurverloop... 2 2.3. Aan- en uitzetten van

### BASIC TECHNICAL SPECIFICATIONS GENERAL SPECIFICATIONS

NEW GENERATION INSTRUMENT FOR SAFETY TESTS ON ELECTRIC PANEL BOARDS, EQUIPMENT AND POWER LINES IN COMPLIANCE WITH IEC/EN60204-1 AND IEC/EN61439-1 GUIDELINES >>> towards the ENSURE THE SAFETY FULLtest3

### EUROsoft. Software HANDLEIDING

EUROsoft Software HANDLEIDING Handleiding BLAUWE LIJN EUROsoft software INHOUDSOPGAVE 1. INSTALLATIE... 2 1.1. Besturingssysteem... 2 1.2. De installatie (Windows XP)... 2 1.3. De installatie ongedaan

### AMP-300-EUR-serie. TRMS-stroomtangen voor motoronderhoud De moderne evolutie van de professionele stroomtang. Kenmerken van de.

AMP-300--serie TRMS-stroomtangen voor motoronderhoud De moderne evolutie van de professionele stroomtang De AMP-300--serie True-RMS-stroomtangen van Beha-Amprobe biedt een compleet scala van meetfuncties

### Artificial Intelligence.

13 P Q A 8 2 0 Artificial Intelligence. Thanks to the creation of App HTanalysis it is possible to interface HT last generation instruments with tablets and smartphones. HTanalysis is a professional software

### AMP-200-EUR-serie TRMS-stroomtangen De moderne evolutie van de professionele stroomtang

AMP-200-EUR-serie TRMS-stroomtangen De moderne evolutie van de professionele stroomtang De 600A-TRMS-stroomtangen AMP-210-EUR en AMP-220-EUR van Beha-Amprobe bieden een compleet scala van meetfuncties

### MERCURY Rel of 19/04/18

TRMS Thermal Multimeter Pag 1 di 4 1. ELECTRICAL SPECIFICATIONS Accuracy calculated as [ reading + (num dgt * resolution)] ta 18 C 28 C,

### Onze producten en aanbiedingen

Onze producten en aanbiedingen Voorjaar/zomer 01-03-2017-30-06-2017 Speciale starterset 2100-Alpha spannings- en doorbeltester IR-710-EUR infraroodthermometer AM-500-EUR digitale multimeter Beha-Amprobe

### DE FLUKE 805 TRILLINGSMETER

DE FLUKE 805 TRILLINGSMETER Vergeet eetet de pennen. Gebruik een METER. De Fluke 805 is geen trillingsmeetpen; het is een meter. Een meter van Fluke die de verwarring bij het meten van trillingen elimineert,

### Correlator for the computer-assisted

Correlator for the computer-assisted SeCorr 08 detection of leaks SeCorr 08 The handy and lightweight correlator for effortless work and listening to signals on the receiver suitable for use in all weather.

### HANDLEIDING Airflow LCA301 vleugelrad anemometer

HANDLEIDING Airflow LCA301 vleugelrad anemometer INHOUD 1. ALGEMENE BESCHRIJVING... 2 2. VEILIGHEID... 2 3. OVER DE LCA301... 2 3.1 Het plaatsen van de batterij... 2 3.2 LCD Display... 2 3.3 Auto Shut

### HANDLEIDING BLAUWE LIJN EUROsoft software

HANDLEIDING BLAUWE LIJN EUROsoft software INHOUD 1. INSTALLATIE... 2 1.1. De installatie... 2 1.2. De installatie ongedaan maken... 2 2. PROGRAMMABESCHRIJVING... 3 2.1. Bedieningspaneel... 3 2.1.1. Display

### Fluke Connect - modules Technische gegevens

Fluke Connect - modules Technische gegevens De flexibiliteit om uw eigen wireless systeem te bouwen zoals u het wilt wanneer u het wilt. De Fluke 3000 FC wireless meetinstrumenten vormen een team en de

### Fluke / Draagbare-apparatentesters

Fluke 6200-2/6500-2 Draagbare-apparatentesters Technische gegevens Voer iedere dag meer tests uit De compacte lichtgewichtoplossing die werkt met één druk op een toets. De Fluke 6200-2 en 6500-2 PAT-testers

### Gas Detection Systems

Gas Detection Systems The Invisible Danger Safety devices and protective measures against hazardous gases and vapours are becoming more and more important. Increasing appreciation of the problem combined

### MIT200-gamma Digitale/Analoge Isolatieweerstand- en Continuïteitstesters

Digitale/Analoge Isolatieweerstand- en Continuïteitstesters MIT200-gamma Digitale/Analoge Isolatieweerstand- en Continuïteitstesters n Testen van isolatieweerstand tot 1000 MΩ n Continuïteitstesten bij

### VARIOTEC 460 Tracergas. The specialist for leak detection with tracer gas and hydrogen

The specialist for leak detection with tracer gas and hydrogen Rely on precision and safety The was developed especially for leak detection on underground pipes by using tracer gas. It is characterised

### FerroTec FT 10. Magnetometer for locating concealed objects fast convenient precise

FerroTec FT 10 Magnetometer for locating concealed objects fast convenient precise FerroTec FT 10 fast convenient precise Easy location of ferromagnetic objects The FerroTec FT 10 helps you safely and

### PROTÉGÉ ZM ENKELGASMONITOR

PROTÉGÉ ZM ENKELGASMONITOR HET BESCHERMEN VAN UW PERSONEEL WORDT STEEDS MAKKELIJKER. PROTÉGÉ ZM ENKELGASMONITOR Scott Safety kondigt de introductie aan van de Protégé ZM gasmonitor; een makkelijk te gebruiken,

### UNIQUE MULTIFUNCTION INSTRUMENT FOR TESTING SAFETY ON ELECTRICAL INSTALLATIONS A NEW WORKING CONCEPT BY HT...

UNIQUE MULTIFUNCTION INSTRUMENT FOR TESTING SAFETY ON ELECTRICAL INSTALLATIONS A NEW WORKING CONCEPT BY HT... MacrotestG3 represents the ideal fusion between HT 30 years experience and the numberless opportunities

### Fluke 370 FC-serie wireless True-RMS-AC/ DC-stroomtangen

TECHNISCHE GEGEVENS Fluke 370 FC-serie wireless True-RMS-AC/ DC-stroomtangen LOGGEN VAN METINGEN EN TRENDREGISTRATIE Resultaten wireless zenden via de Fluke Connect - app voor het uitvoeren van metingen

### HANDLEIDING BLAUWE LIJN GSP-3 (flex) gaslekzoekers

HANDLEIDING BLAUWE LIJN GSP-3 (flex) gaslekzoekers GSP 3 fl ex GSP 3 HANDLEIDING BLAUWE LIJN GSP-3 (flex) gaslekzoekers INHOUD 1. OMSCHRIJVING VAN HET TOESTEL... 2 2. WERKING VAN HET TOESTEL... 2 2.1 Inschakelen

### HANDLEIDING BLAUWE LIJN GSP-1 gaslekzoeker

HANDLEIDING BLAUWE LIJN GSP-1 gaslekzoeker INHOUD 1. OMSCHRIJVING APPARAAT... 2 2. TECHNISCHE GEGEVENS... 2 3. MEETWERKING... 3 4. INSTELLING VAN AKOESTISCH ALARM... 3 5. OPLADEN VAN DE ACCU... 4 6. VEILIGHEIDSASPECTEN...

### VT04, VT04A en VT02 visuele infraroodthermometers

VT04, VT04A en VT02 visuele infraroodthermometers Technische gegevens Problemen onmiddellijk opsporen! Elimineer de noodzaak om meerdere temperatuurmetingen uit te voeren. Visuele infraroodthermometers

### Fluke 287 en 289 digitale multimeters

Fluke 287 en 289 digitale multimeters Onze meest geavanceerde multimeters met logfunctie kunnen u waardevolle tijd besparen Vind kleine oorzaken voordat ze grote problemen opleveren De Fluke 287 en 289

### Fluke 1660-serie Multifunctionele installatietesters

TECHNISCHE GEGEVENS Fluke 1660-serie Multifunctionele installatietesters De nieuwe Fluke 1660-serie installatietesters zijn de enige installatietesters die schade aan aangesloten apparaten helpen voorkomen

### 1507/1503. Insulation Testers. Technical Data. Features and benefits:

1507/1503 Insulation Testers Technical Data The Fluke 1507 and 1503 Insulation Testers are compact, rugged, reliable, and easy to use. With their multiple test voltages, they are ideal for many troubleshooting,

### Fluke 805 trillingsmeter

Fluke 805 trillingsmeter Technische gegevens De betrouwbare, reproduceerbare, nauwkeurige methode om lagers en algemene trillingsniveaus te controleren. Neem vol vertrouwen go/no-go -beslissingen. De Fluke

### Air Volume Instruments

Features and Benefits Model PH731 (Accessories shown on following page) ProHood Capture Hood Model PH731 The PH731 ProHood Capture Hood is a multipurpose electronic air balancing instrument primarily used

### UtiliTrac. The standard in pipe and cable detection

The standard in pipe and cable detection Design meets functionality The UtiliTrac sets new standards in locating underground pipes and cables. Ist ergonomic design and intuitive trigger controlled functions

### SePem. Noise loggers for monitoring water pipe networks sturdy convenient reliable

SePem Noise loggers for monitoring water pipe networks sturdy convenient reliable SePem sturdy convenient reliable Detect water losses early Leaks in water pipe networks can result in significant water

### ExDetector IRCO2. ExDetector IRHC. Data Sheet. Gas Detection and Warning Systems

ExDetector IRCO2 ExDetector IRHC Gas Detection and Warning Systems Installation / Set-up In combination with a controller system, gas detectors in the ExDetector IR range haven the following functions:

### Biogas Analyzer GMC Biogas 08

Biogas Analyzer Gas Detection Systems Bieler + Lang Established in 1967, Bieler + Lang is your competent partner for gas measuring and detection systems. Since 1970 Bieler + Lang is producing gas measuring

### SPX Model A-360 Azimuth Antenna Rotor Model 1 & 2

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com Model A-360 Azimuth

### Fluke 279 FC thermische multimeter

TECHNISCHE GEGEVENS Fluke 279 FC thermische multimeter CAMERA Ingebouwde warmtebeeldcamera DISPLAY Het LCD-kleurenscherm biedt heldere en scherpe meetwaarden iflex Breid uw meetmogelijkheden uit. U kunt

### I.S.T.C. Intelligent Saving Temperature Controler

MATEN & INFORMATIE I.S.T.C. Intelligent Saving Temperature Controler Deze unieke modulerende zender, als enige ter wereld, verlaagt het energieverbruik aanzienlijk. Het werkt in combinatie met de energy

### Fluke 805 FC Trillingsmeter

Fluke 805 FC Trillingsmeter Technische gegevens Nu compatibel met Fluke Connect Mobile App De betrouwbare, reproduceerbare, nauwkeurige methode om lagers en algemene trillingsniveaus te controleren. Neem

### die werkt met één druk op de toets

De compacte lichtgewichtoplossing die werkt met één druk op de toets Testen van draagbare apparaten De Fluke 6000-2 PAT-testers Fluke 6200-2 en 6500-2 Test meer apparaten per dag De Fluke 6200-2 en 6500-2

### FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE. Toets Inleiding Kansrekening 1 8 februari 2010

FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE Toets Inleiding Kansrekening 1 8 februari 2010 Voeg aan het antwoord van een opgave altijd het bewijs, de berekening of de argumentatie toe. Als je een onderdeel

### HANDLEIDING BLAUWE LIJN Afpersset tot 1 bar en Afpersset tot 10 bar

HANDLEIDING BLAUWE LIJN Afpersset tot 1 bar en Afpersset tot 10 bar INHOUD 1. DE DRUKADAPTER... 2 1.1. Functionaliteit... 2 1.2. Montage van de drukadapter op de S2600- en S4600-serie drukmeters... 2 2.

Quick start guide Powerbank MI 16.000 Mah Follow Fast 2016 - All rights reserved. Page 1 ENGLISH The Mi 16000 Power Bank is a very good backup option for those on the move. It can keep you going for days

### VARIOTEC 460 EX 450 EX 400 EX. Cost-effective gas detector for distribution networks

VARIOTEC 460 450 400 Cost-effective gas detector for distribution networks The entry into professional gas measurement The VARIOTEC 4x0 range gives network technicians an efficient and cost effective means

### 600A Clamp-On Multimeter with Dual Display

ACD-14 PLUS 600A Clamp-On Multimeter with Dual Display The ACD-14 PLUS is the professionals choice for both HVAC and Electrical applications due to its complete set of measuring features. Its dual displays

### Introductie in flowcharts

Introductie in flowcharts Flow Charts Een flow chart kan gebruikt worden om: Processen definieren en analyseren. Een beeld vormen van een proces voor analyse, discussie of communicatie. Het definieren,

### Groene industrie POWER QUALITY POWER AND SIGNAL QUALITY. Phoenix Contact B.V. 2015-2016

Groene industrie Phoenix Contact B.V. 2015-2016 POWER AND SIGNAL QUALITY POWER QUALITY 24V PWR ERR 24V 0V 0V OUT I GND GND GND ON OFF 1 2 1A 5A NC PACT RCP- Rogowskisp Eerst Power Quality inzichtelijk

### PIR DC-SWITCH. DC Passive infra-red Detector. Model No. PDS-10 GEBRUIKSAANWIJZING/INSTRUCTION MANUAL

PIR DC-SWITCH DC Passive infra-red Detector Model No. PDS-10 GEBRUIKSAANWIJZING/INSTRUCTION MANUAL Please read this manual before operating your DETECTOR PIR DC-Switch (PDS-10) De PDS-10 is een beweging

### HANDLEIDING - ACTIEVE MOTORKRAAN

M A N U A L HANDLEIDING - ACTIEVE MOTORKRAAN MANUAL - ACTIVE MOTOR VALVE Model E710877 E710878 E710856 E710972 E710973 www.tasseron.nl Inhoud / Content NEDERLANDS Hoofdstuk Pagina NL 1 ALGEMEEN 2 NL 1.1

### Fluke-meetinstrumenten met ATEX-certificering voor explosiegevaarlijke omgevingen

Fluke-meetinstrumenten met ATEX-certificering voor explosiegevaarlijke omgevingen Intrinsiek veilige multimeteren proceskalibrators Conform aan ATEX Intrinsiek veilige Fluke-instrumenten voor metingen

### Professional PC correlation SeCorr 300 Perfect results thanks to fully digital technology

Professional PC correlation SeCorr 300 Perfect results thanks to fully digital technology SeCorr 300 The principle SEWERIN has a long tradition of producing systems for locating damages in pipe networks

### Combined stationary measuring device

Combined stationary measuring device with mobile gas meter Determines the composition and volume of biogas, sewer gas and landfill gas All plant operators producing electricity from biogas, sewer gas or

### NIEUW. Extra functionaliteit, sneller testen, en nog steeds zeer robuust

Extra functionaliteit, sneller testen, en nog steeds zeer robuust Met de Fluke 1650-serie multifunctionele testers kunt u testen conform NEN 1010, NEN 3140 en EN61557 NIEUW Geschikt voor Type B aardlekschakelaar

### SPX AX-400 Azimuth & Elevation antenna rotator Model 1 & 2

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com SPX AX-400 Azimuth

### WeZoWe - Werkgroep Zonnepanelen Westeinde. Het adres voor al uw installatiewerk!!

WeZoWe - Werkgroep Zonnepanelen Westeinde Het adres voor al uw installatiewerk!! Inhoud: o Over ons o Advies o Montage o Waarom Zonderland o Gerealiseerde projecten o Garantie o Prijzen o Stand o Advertentie

### HANDLEIDING BLAUWE LIJN S K(D) temperatuurmeters

HANDLEIDING BLAUWE LIJN S2600-12K(D) temperatuurmeters S2600-12KD S2600-12K INHOUD 1. BELANGRIJK... 2 2. NEN-EN 50379:2004... 2 3. BEDIENING EN WERKWIJZE... 2 3.1. Batterij... 2 3.2. Temperatuurverloop...

### BEKIJKEN. OPSLAAN. DELEN. Alle feiten, meteen op locatie.

BEKIJKEN. OPSLAAN. DELEN. Alle feiten, meteen op locatie. Introductie van het grootste systeem van met elkaar verbonden test- en meetinstrumenten ter wereld. Introductie van het grootste systeem van met

### Classification of triangles

Classification of triangles A triangle is a geometrical shape that is formed when 3 non-collinear points are joined. The joining line segments are the sides of the triangle. The angles in between the sides

### Technologies for leak detection. AQUAPHON. Electro-acoustic water leak detection

Technologies for leak detection. AQUAPHON AQUAPHON A 00 receiver Electro-acoustic water leak detection Innovative technologies The principle When pressure pipelines leak, water gushes out of the crack

### STATIONAIRE MANOMETERS MANOMÈTRES STATIONNAIRES

Algemene informatie / Information générale Stationaire manometers kunnen worden voorzien van uw bedrijfslogo. Prijzen op aanvraag. KWS (kalibratie) is mogelijk op stationaire manometers tot 60 bar (code

### BLAUWE LIJN S K(D) temperatuurmeters

BLAUWE LIJN S2600-12K(D) temperatuurmeters Displayverlichting Geschikt voor alle K-type thermokoppels Eén of twee temperatuurkanalen Meten van temperatuurverschillen (S2600-12KD) Holdfunctie MIN- / MAX-functie

### SPID Azimuth antenna rotator Type: BIG-RAK

SPID Azimuth antenna rotator Type: BIG-RAK Specifications: SPID BIG RAK PWM Azimuth Rotator BIG RAK @ 12V BIG RAK @ 18V Turning torque in-lbs/nm 2,800 in lb/315 Nm 3,600 in lb/403 Nm Brake Torque in lbs/nm

### SPL D2 MKII GEBRUIKSAANWIJZING USER MANUAL

SPL D2 MKII GEBRUIKSAANWIJZING USER MANUAL Spl-D2 Next generation Inleiding De SPL5-D2 unit is een geluidsdrukmeter die gekoppeld kan worden aan de SPL5. Het apparaat kan ook als losse geluidsdrukmeter

### SF6 GASCHECK 6000 SF6 LEAK DETECTOR LOW COST, EFFECTIVE DETECTION OF SF6 LEAKS. ionscience.com Unrivalled Gas Detection.

SF6 GASCHECK 6000 SF6 LEAK DETECTOR LOW COST, EFFECTIVE DETECTION OF SF6 LEAKS. THE SF6 GASCHECK 6000 IS A LOW COST, PORTABLE DETECTOR FOR THE FAST DETECTION OF SF6 LEAKS. Fast, accurate leak detection

### Fluke 3000 FC serie meetinstrumenten Technische gegevens

Fluke 3000 FC serie meetinstrumenten Technische gegevens Nu compatibel met Fluke Connect Mobile App Begin vandaag nog met het bouwen van het systeem van meetinstrumenten van morgen, met de digitale multimeter

### SPID Azimuth antenna rotator Type: RAK

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com SPID Azimuth antenna

### AT-7000-EUR geavanceerde kabelzoeker

AT-7000-EUR geavanceerde kabelzoeker Kabels zoeken opnieuw uitgevonden Nauwkeurige resultaten binnen enkele minuten met nieuwe functies en technologieën die het zoeken van kabels en het identificeren van

### Hydrosteel 6000. The Hydrosteel 6000 hydrogen flux monitor is the ultimate portable tool for active sour, HF and high temperature corrosion.

Hydrosteel 6000 Portable, rapid identification of active corrosion & hydrogen damage hotspots. The Hydrosteel 6000 hydrogen flux monitor is the ultimate portable tool for active sour, HF and high temperature

### Met elke andere Power Quality Analyzer verspilt u slechts energie.

Met elke andere Power Quality Analyzer verspilt u slechts energie. Fluke 430-serie II Power Quality- en energieanalyzers Modellen uit de Fluke 430-serie II Fluke 434, serie II energieanalyzer De Fluke

### FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE

FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE Tentamen Analyse 6 januari 203, duur 3 uur. Voeg aan het antwoord van een opgave altijd het bewijs, de berekening of de argumentatie toe. Als je een onderdeel

### T90/T110/T130/T150 Spannings- en doorbeltesters

T90/T110/T130/T150 Spannings- en doorbeltesters Robuuste, hoogwaardige testers geven snel testresultaten, op de manier die u het beste uitkomt Alle elektriciens zouden een tweepolige tester moeten hebben.

### Artificial Intelligence.

5 M A C R O T E S T G 3 Artificial Intelligence. Thanks to the creation of App HTanalysis it is possible to interface HT last generation instruments with tablets and smartphones. HTanalysis is a professional

### liniled Cast Joint liniled Gietmof liniled Castjoint

liniled Cast Joint liniled Gietmof liniled is een hoogwaardige, flexibele LED strip. Deze flexibiliteit zorgt voor een zeer brede toepasbaarheid. liniled kan zowel binnen als buiten in functionele en decoratieve

### Gas Measuring and Alarm Systems

Exmonitor Gasmonitor Gas Measuring and Alarm Systems Application/Structure The sensors of the series Exmonitor/Gasmonitor used in combination with the analysis systems have the following functions: Measuring

### L.Net s88sd16-n aansluitingen en programmering.

De L.Net s88sd16-n wordt via één van de L.Net aansluitingen aangesloten op de LocoNet aansluiting van de centrale, bij een Intellibox of Twin-Center is dat de LocoNet-T aansluiting. L.Net s88sd16-n aansluitingen

### Installatie-/Gebruikershandleiding. FluoMini V2. Optical Temperature Sensor.

Installatie-/Gebruikershandleiding FluoMini T Optical Temperature Sensor www.sendot.nl Inhoudsopgave 1. Algemeen... 3 2. Specificaties... 4 3. Installatie... 4 4. Onderhoud/Kalibratie... 5 1. Algemeen

### Fluke 170 Series True-rms Digital Multimeters Detailed Specifications

Fluke 170 Series True-rms Digital Multimeters Detailed Specifications For all detailed specifications Accuracy is specified for 1 year after calibration, at operating temperatures of 18 C to 28 C, with

### Multitec BioControl system BioControl 2. The entry-level model for individual process optimisation

Multitec BioControl system BioControl 2 The entry-level model for individual process optimisation BioControl 2: user-friendly, flexible, efficient The entry-level model for automated process optimisation

### TiS10, TiS20, TiS40, TiS45, TiS50, TiS55, TiS60, TiS65 Warmtebeeldcamera s

TECHNISCHE GEGEVENS TiS10, TiS20, TiS40, TiS45, TiS50, TiS55, TiS60, TiS65 Warmtebeeldcamera s Built with Compatibel met de Fluke Connect Mobile App De Fluke Performanceserie Gebouwd voor prestaties, met

### FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE

FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE Tentamen Analyse 8 december 203, duur 3 uur. Voeg aan het antwoord van een opgave altijd het bewijs, de berekening of de argumentatie toe. Als jeeen onderdeel

### MIT400/2 CAT IV Insulation testers

MIT400/2 CAT IV Insulation testers MIT400/2 CAT IV Insulation testers Designed for Electrical and Industrial testing Insulation testing up to 1000 V and 200 GΩ range Stabilised insulation test voltage

### ph ORP Temperatuur Geleidbaarheid TDS Opgeloste Zuurstof Salinity Weerstand

ph ORP Temperatuur Geleidbaarheid TDS Opgeloste Zuurstof Salinity Weerstand De Perfecte Combinatie Dit Chinese symbool betekend Dubbel geluk, de combinatie tussen twee perfecte llementen Dit wordt gebruikt

### L.Net s88sd16-n aansluitingen en programmering.

De L.Net s88sd16-n wordt via één van de L.Net aansluitingen aangesloten op de LocoNet aansluiting van de centrale, bij een Intellibox of Twin-Center is dat de LocoNet-T aansluiting. L.Net s88sd16-n aansluitingen

### FlowTemp ST Digitale waterhoeveelheid- en temperatuurmeter

FlowTemp ST Digitale waterhoeveelheid- en temperatuurmeter Handleiding BLAUWE LIJN FlowTemp ST digitale waterhoeveelheid- en temperatuurmeter INHOUDSOPGAVE 1. OVER DEZE HANDLEIDING... 2 2. VEILIGHEID...

### Het is geen open boek tentamen. Wel mag gebruik gemaakt worden van een A4- tje met eigen aantekeningen.

Examen ET1205-D1 Elektronische Circuits deel 1, 5 April 2011, 9-12 uur Het is geen open boek tentamen. Wel mag gebruik gemaakt worden van een A4- tje met eigen aantekeningen. Indien, bij het multiple choice