# MPP300 Rel /03/11

Save this PDF as:

Maat: px
Weergave met pagina beginnen:

## Transcriptie

1 Accessory for checks on PV plants with multi MPPT Pag 1 of 4 1. NOTES ON MPPT (Maximum Power Point Tracker) Solar irradiation on a surface such as the surface of a photovoltaic system has extremely variable characteristics, since it depends on the position of the sun with respect to the surface and on atmospheric conditions (typically, on the presence of clouds). A photovoltaic module presents, for different solar irradiation values, and for different temperature values, a range of characteristic curves of the type shown in the following figure. In particular, the figure shows three I-V curves (in bold) which correspond to three values (1000, 800, 600W/m 2 ) of solar irradiation On each characteristic curve there is one single point in which the power transfer towards a hypothetical charge supplied by the photovoltaic module is maximized. The maximum power point corresponds to the voltage-current pair for which the product V*I is maximum, where V is the value of voltage at the module s terminals and I is the current which runs in the circuit obtained by closing the module on a hypothetical charge. With reference to the figure above, the product V*I is represented, for the three solar irradiation values mentioned above, through the three curves in thinner lines. The figure shows that, as stated above, these curves only have one single maximum point. For example, for 1000W/m 2, the maximum power point corresponds to a voltage value of approx. 36V and to a current value of approx. 5.5A. Obviously, if the power provided by the system is maximized, it is possible to make the most of the system, both in case the system is connected to mains, and in case it is stand-alone. MPPT is an inbuilt device in the inverters. It typically reads the voltage and current values at any instant, calculates their product (i.e. the power in Watts) and, by causing small variations in the conversion parameters (duty cycle), it is capable of determining, by comparison, if the photovoltaic module is working in maximum power conditions or not. According to the result, it operates again on the circuit in order to bring the system to an optimal condition. The reason why MPPTs are used is simple: a photovoltaic system without MPPTs may operate anyway. However, with the same solar irradiation, it provides less energy. There are inverters with 1, 2 or also 3 inbuilt MPPTs available on the market. Typically, the inverters with more than one MPPT are used in systems where: The different photovoltaic systems it consists of forcibly have different inclinations or directions. In this way, each single MPPT manages its own photovoltaic field, maximizing its performance for the corresponding irradiation and temperature characteristics (without being influenced by the other photovoltaic fields) A greater service continuity is sought. With more MPPTs it is possible to put one single photovoltaic field out of service, while the others continue producing energy towards the remaining MPPTs.

2 Accessory for checks on PV plants with multi MPPT Pag 2 of 4 2. DESCRIPTION OF MPP300 ACCESSORY The MPP300 model has been designed as an exclusive accessory of a Master SOLAR300N and SOLAR I-V instruments for the purpose of carrying out testing simultaneous operations on single-phase and three-phase PV systems with up to 3 Multi MPPT device Together with a Master instrument (see enclosed principle scheme) MPP300 is the ideal solution for testing and analyzing the possible problems linked to possible low efficiency values of photovoltaic systems The instrument has the following features: Testing of PV systems with single/multi-mppt inverter - single/three-phase AC output Using with Master instruments SOLAR300N and SOLAR I-V Measurement of 3 DC voltages and currents Measurement of DC string power and total DC power Measurement of 3 AC TRMS voltages and currents Measurement of total AC power Measurement of irradiation [W/m 2 ] by means of a reference cell connected to unit SOLAR-02 Measurement of panel and environmental temperature by means of probe PT300N connected to SOLAR-02 Parameter recording of a PV system with 5s to 60min programmable IP Operations with LED indication Internal memory for data saving RF interface for transferring the data to the SOLAR I-V instrument and SOLAR-02 remote unit USB interface for transferring the data to the SOLAR300N instrument

3 Accessory for checks on PV plants with multi MPPT Pag 3 of 4 3. TECHNICAL SPECIFICATIONS (*) Accuracy is indicated as [%reading + (no. of digits) * resolution] at 23 C ± 5 C, <80%HR DC Voltage Range (V) Resolution (V) Accuracy (0.5 rdg + 2dgt) AC TRMS Voltage Phase-Neutral Single/Three phase systems Range (V) Frequency (Hz) Resolution (V) Accuracy (0.5 rdg + 2cifre) Max. crest factor: 1.5 AC TRMS Voltage Phase-Phase Range (V) Frequency (Hz) Resolution (V) Accuracy (0.7 rdg + 2dgt) Max. crest factor: 1.5 DC Current by means of external clamp transducer STD type Range (mv) Resolution (mv) Accuracy 5.0mV 319.9mV (0.5 rdg %FS) 0.1mV 10V 320.0mV 999.9mV 0.5 rdg Current values corresponding to a voltage < 5mV are zeroed AC TRMS Current by means of external clamp transducer STD type 5.0mV 219.9mV (0.5 rdg %FS) mV 220.0mV 999.9mV 0.5 rdg Max. crest factor: 1.5 ; Current values corresponding to a voltage < 5mV are zeroed 10V AC TRMS current (by means of external clamp transducer of FLEX 8.5µV/A FS 100A) mV Hz 0.001mV (0.5%rdg + 7dgt) 10V Max. crest factor: 1.5 ; Current values < 1A are zeroed AC TRMS current (by means of external clamp transducer of FLEX 8.5µV/A FS 1000A) mV Hz 0.01mV (0.5%rdg + 15dgt) 10V Max. crest factor: 1.5 ; Current values < 5A are zeroed DC Power (Vmis > 150V) Parameter FS clamp (A) Range [W] Resolution [W] Accuracy 0.000k 9.999k 0.001k 1< FS k 99.99k 0,01k (0.7 rdg+3dgt) DC POWER 10< FS k 100.0k 999.9k 0.0k 999.9k 100< FS k 9999k Vmis = voltage at which power is measured; Imis = measured current 0.01k 1k (Imis < 10%FS) (0.7 rdg) (Imis 10%FS)

4 Accessory for checks on PV plants with multi MPPT Pag 4 of 4 AC Power (Vmis > 200V, PF=1) Parameter FS clamp (A) Range [W] Resolution [W] Accuracy 0.000k 9.999k 0.001k 1< FS k 99.99k 0,01k (0.7 rdg+3dgt) AC POWER 10< FS k 100.0k 999.9k 0.0k 999.9k 200< FS k 9999k Vmis = voltage at which power is measured; Imis = measured current 4. GENERAL SPECIFICATIONS 0.01k 1k (Imis < 10%FS) (0.7 rdg) (Imis 10%FS) POWER SUPPLY: Internal power supply: Battery duration: External power supply: Internal rechargeable Li-ION battery (3.7V, 1900mAh) > 3 hours AC/DC adapter V 50/60Hz / 5VDC CHARACTERISTICS OF RADIO MODULE Frequency range: GHz R&TTE category: Class 1 Max transmission power: 30µW Max RF connection distance: 1m MEMORY AND EXTERNAL INTERFACE Memory capacity: 2Mbytes Integration Period (IP): 5,10,30,60,120,300,600,900,1800,3600s RF interface: connection SOLAR I-V and SOLAR-02 units USB interface: connection SOLAR300N unit Battery duration (with SOLAR-02): approx. 1.5 hours PI = 5s) ; approx. 8 days PI = 600s) MECHANICAL CHARACTERISTICS Dimensions (LxWxH): Weight (battery included): Protection index: 300 x 265 x 140 mm 1.2 kg IP40 ENVIRONMENTAL CONDITION FOR USE Reference temperature: 23 C 5 C Operating temperature: 0 40 C Allowable relative humidity: <80%hr Storage temperature: C Storage humidity: <80%HR REFERENCE STANDARDS Safety: IEC/EN Safety of measuring accessories: IEC / EN Insulation: double insulation Pollution level: 2 Measurement category: CAT III 1000V DC, Max 1000V between DC inputs CAT IV 300V AC to ground, Max 600V between AC inputs Max operating altitude: 2000m This instrument satisfies the requirements of Low Voltage Directive 2006/95/EC (LVD) and of EMC Directive 2004/108/EC (*) Technical specifications can be modified without preliminary advise

5 EURO-INDEX Service Onderhoud en kalibratie van meetinstrumenten... EURO-INDEX beschikt over een bijzonder modern service- en kalibratielaboratorium. Hier worden de meetinstrumenten uit het assortiment preventief onderhouden, gerepareerd, gekalibreerd en indien nodig gejusteerd. Het service- en kalibratielaboratorium van EURO-INDEX is verdeeld in verschillende disciplines, gebaseerd op het soort meetinstrument en de gemeten grootheden. Druk Gasanalyse Temperatuur (inclusief infrarood temperatuurmeting en thermografie) Elektrische grootheden Gasdetectie Luchtsnelheid en luchthoeveelheid Waarom een kalibratiecertificaat?... Een kalibratiecertificaat vermeldt hoeveel een meetinstrument afwijkt ten opzichte van onze, naar (inter)nationale standaarden herleidbare, kalibratiemiddelen. Bij de meetresultaten op het certificaat wordt tevens vermeld of het meetinstrument voldoet aan de specificaties die door de fabrikant zijn opgegeven. Zonder kalibratiecertificaat kunt u er vanuit gaan dat de meter voldoet aan de fabrieksspecificaties, maar aantonen kunt u dit niet. Een testcertificaat van de fabrikant is te beknopt om de lineairiteit aan te tonen en is niet geregistreerd op naam (wat wel degelijk een vereiste is). SPECIMENEN SPECIMEN KWS... KWS is een uniek servicesysteem voor uw meetinstrumenten met periodiek onderhoud en kalibratie. Veel zaken worden voor u geregeld, zodat u zonder zorgen gebruik kunt maken van uw meetinstrumenten. De prijs staat vast voor de levensduur van het instrument (mits de KWS behandeling volgens herkalibratieadvies periodiek wordt uitgevoerd in het EURO-INDEX kalibratielaboratorium) Geen arbeidsloon bij de KWS behandeling Kalibratie voor justage (voorkalibratie) indien mogelijk Indien nodig justage en (na)kalibratie Reparatie en preventief onderhoud Gratis oproep met het advies voor herkalibratie Controle op functionaliteit van het instrument Vijf jaar historie voor alle gegevens 10% korting op onderdelen Serienummerregistratie Franco retourlevering RvA accreditatie... Het kalibratielaboratorium van EURO-INDEX beschikt sinds 21 augustus 1997 over een RvA accreditatie naar NEN-EN-ISO/IEC Deze accreditatie geldt voor verschillende grootheden, zoals gespecificeerd in de scope bij accreditatienummer K105 op Test- en meetinstrumenten voor grootheden die deel uitmaken van de gespecificeerde scope, kunnen worden voorzien van een RvA kalibratiecertificaat. De metingen worden uitgevoerd met standaarden waarvan de herleidbaarheid naar (inter)nationale standaarden, ten overstaan van de Raad voor Accreditatie, is aangetoond. In het Multilateral Agreement zijn de meeste Europese landen overeengekomen elkaars accreditaties te accepteren. Hierdoor is een RvA kalibratiecertificaat internationaal geaccepteerd. Bovendien wordt op een RvA kalibratiecertificaat de meetonzekerheid van de gerapporteerde meetresultaten vermeld. Verhuur van meetinstrumenten... EURO-INDEX biedt een assortiment meetinstrumenten te huur aan. Na deskundig advies van onze productspecialisten, wordt bepaald welk instrument u nodig heeft voor uw specifieke werkzaamheden. De instrumenten worden compleet met accessoires geleverd, inclusief herleidbaar kalibratiecertificaat. Wijzigingen voorbehouden EURO-INDEX NL NEDERLAND Rivium 2e straat LG Capelle a/d IJssel T: F: Leuvensesteenweg 607 BELGIË 1930 Zaventem T: (0) F: (0)

### SOLAR I-V Rel. 1.01 20/07/10

PV single phase certifier and I-V curve checker Pag 1 of 5 1. SOLAR I-V MAIN FEATURES SOLAR I-V: use as PV certifier SOLAR I-V performs all tests on Single phase PV plants by using of SOLAR-02 remote unit

### SOLAR I-Vw Rel. 1.03 06/10/15

PV single phase certifier and I-V curve checker up to 15A Pag 1 of 5 1. SOLAR I-Vw MAIN FEATURES SOLAR I-Vw: use as PV certifier SOLAR I-Vw performs all tests on Single phase PV plants by using of SOLAR-02

### SOLAR300N Rel. 1.03 21/01/10

Photovoltaic installation certifier + professional network analyzer Pag 1 of 5 1. SOLAR300N MAIN FEATURES SOLAR300N performs all tests on PV plants by using of SOLAR-02 remote unit which, after a preliminary

### SECUTEST BASE VEILIGHEIDS BASE10 DE NIEUWE SECUTEST VEILIG EN EFFICIËNT TESTEN VAN ELEKTRISCHE ARBEIDSMIDDELEN

DE NIEUWE SECUTEST SECUTEST BASE VEILIGHEIDS BASE10 VEILIG EN EFFICIËNT TESTEN VAN ELEKTRISCHE ARBEIDSMIDDELEN Elektrische apparaten volgens NEN3140 Incl. verlengkabels en (P) RCD s Medische apparaten

### Klein formaat, grote prestaties AMP-25-EUR AC/DC TRMS Mini-Stroomtang

Herfst/winter 2015 Periode: 01.09.2015 31.12.2015 Informatie over de nieuwste producten en acties van BehaAmprobe. Klein formaat, grote prestaties AMP25EUR AC/DC TRMS MiniStroomtang Kabels zoeken opnieuw

### Database Memory HANDLEIDING

Database Memory HANDLEIDING Handleiding BLAUWE LIJN Database Memory INHOUDSOPGAVE 1. OVER DEZE HANDLEIDING... 2 2. KLANTGEGEVENS... 2 2.1. Invoeren van klantgegevens... 2 2.2. Klantgevens aanmaken/bewerken...

### STATIONAIRE MANOMETERS MANOMÈTRES STATIONNAIRES

Algemene informatie / Information générale Stationaire manometers kunnen worden voorzien van uw bedrijfslogo. Prijzen op aanvraag. KWS (kalibratie) is mogelijk op stationaire manometers tot 25 bar (code

### BASIC TECHNICAL SPECIFICATIONS GENERAL SPECIFICATIONS

NEW GENERATION INSTRUMENT FOR SAFETY TESTS ON ELECTRIC PANEL BOARDS, EQUIPMENT AND POWER LINES IN COMPLIANCE WITH IEC/EN60204-1 AND IEC/EN61439-1 GUIDELINES >>> towards the ENSURE THE SAFETY FULLtest3

### EUROsoft. Software HANDLEIDING

EUROsoft Software HANDLEIDING Handleiding BLAUWE LIJN EUROsoft software INHOUDSOPGAVE 1. INSTALLATIE... 2 1.1. Besturingssysteem... 2 1.2. De installatie (Windows XP)... 2 1.3. De installatie ongedaan

### AMP-200-EUR-serie TRMS-stroomtangen De moderne evolutie van de professionele stroomtang

AMP-200-EUR-serie TRMS-stroomtangen De moderne evolutie van de professionele stroomtang De 600A-TRMS-stroomtangen AMP-210-EUR en AMP-220-EUR van Beha-Amprobe bieden een compleet scala van meetfuncties

### Artificial Intelligence.

13 P Q A 8 2 0 Artificial Intelligence. Thanks to the creation of App HTanalysis it is possible to interface HT last generation instruments with tablets and smartphones. HTanalysis is a professional software

### DE FLUKE 805 TRILLINGSMETER

DE FLUKE 805 TRILLINGSMETER Vergeet eetet de pennen. Gebruik een METER. De Fluke 805 is geen trillingsmeetpen; het is een meter. Een meter van Fluke die de verwarring bij het meten van trillingen elimineert,

### HANDLEIDING BLAUWE LIJN EUROsoft software

HANDLEIDING BLAUWE LIJN EUROsoft software INHOUD 1. INSTALLATIE... 2 1.1. De installatie... 2 1.2. De installatie ongedaan maken... 2 2. PROGRAMMABESCHRIJVING... 3 2.1. Bedieningspaneel... 3 2.1.1. Display

### HANDLEIDING Airflow LCA301 vleugelrad anemometer

HANDLEIDING Airflow LCA301 vleugelrad anemometer INHOUD 1. ALGEMENE BESCHRIJVING... 2 2. VEILIGHEID... 2 3. OVER DE LCA301... 2 3.1 Het plaatsen van de batterij... 2 3.2 LCD Display... 2 3.3 Auto Shut

### Fluke Connect - modules Technische gegevens

Fluke Connect - modules Technische gegevens De flexibiliteit om uw eigen wireless systeem te bouwen zoals u het wilt wanneer u het wilt. De Fluke 3000 FC wireless meetinstrumenten vormen een team en de

### Correlator for the computer-assisted

Correlator for the computer-assisted SeCorr 08 detection of leaks SeCorr 08 The handy and lightweight correlator for effortless work and listening to signals on the receiver suitable for use in all weather.

### Gas Detection Systems

Gas Detection Systems The Invisible Danger Safety devices and protective measures against hazardous gases and vapours are becoming more and more important. Increasing appreciation of the problem combined

### MIT200-gamma Digitale/Analoge Isolatieweerstand- en Continuïteitstesters

Digitale/Analoge Isolatieweerstand- en Continuïteitstesters MIT200-gamma Digitale/Analoge Isolatieweerstand- en Continuïteitstesters n Testen van isolatieweerstand tot 1000 MΩ n Continuïteitstesten bij

### PROTÉGÉ ZM ENKELGASMONITOR

PROTÉGÉ ZM ENKELGASMONITOR HET BESCHERMEN VAN UW PERSONEEL WORDT STEEDS MAKKELIJKER. PROTÉGÉ ZM ENKELGASMONITOR Scott Safety kondigt de introductie aan van de Protégé ZM gasmonitor; een makkelijk te gebruiken,

### UNIQUE MULTIFUNCTION INSTRUMENT FOR TESTING SAFETY ON ELECTRICAL INSTALLATIONS A NEW WORKING CONCEPT BY HT...

UNIQUE MULTIFUNCTION INSTRUMENT FOR TESTING SAFETY ON ELECTRICAL INSTALLATIONS A NEW WORKING CONCEPT BY HT... MacrotestG3 represents the ideal fusion between HT 30 years experience and the numberless opportunities

### FerroTec FT 10. Magnetometer for locating concealed objects fast convenient precise

FerroTec FT 10 Magnetometer for locating concealed objects fast convenient precise FerroTec FT 10 fast convenient precise Easy location of ferromagnetic objects The FerroTec FT 10 helps you safely and

### HANDLEIDING BLAUWE LIJN GSP-1 gaslekzoeker

HANDLEIDING BLAUWE LIJN GSP-1 gaslekzoeker INHOUD 1. OMSCHRIJVING APPARAAT... 2 2. TECHNISCHE GEGEVENS... 2 3. MEETWERKING... 3 4. INSTELLING VAN AKOESTISCH ALARM... 3 5. OPLADEN VAN DE ACCU... 4 6. VEILIGHEIDSASPECTEN...

### Fluke 1660-serie Multifunctionele installatietesters

TECHNISCHE GEGEVENS Fluke 1660-serie Multifunctionele installatietesters De nieuwe Fluke 1660-serie installatietesters zijn de enige installatietesters die schade aan aangesloten apparaten helpen voorkomen

### Fluke 805 trillingsmeter

Fluke 805 trillingsmeter Technische gegevens De betrouwbare, reproduceerbare, nauwkeurige methode om lagers en algemene trillingsniveaus te controleren. Neem vol vertrouwen go/no-go -beslissingen. De Fluke

### die werkt met één druk op de toets

De compacte lichtgewichtoplossing die werkt met één druk op de toets Testen van draagbare apparaten De Fluke 6000-2 PAT-testers Fluke 6200-2 en 6500-2 Test meer apparaten per dag De Fluke 6200-2 en 6500-2

### VARIOTEC 460 EX 450 EX 400 EX. Cost-effective gas detector for distribution networks

VARIOTEC 460 450 400 Cost-effective gas detector for distribution networks The entry into professional gas measurement The VARIOTEC 4x0 range gives network technicians an efficient and cost effective means

### HANDLEIDING BLAUWE LIJN Afpersset tot 1 bar en Afpersset tot 10 bar

HANDLEIDING BLAUWE LIJN Afpersset tot 1 bar en Afpersset tot 10 bar INHOUD 1. DE DRUKADAPTER... 2 1.1. Functionaliteit... 2 1.2. Montage van de drukadapter op de S2600- en S4600-serie drukmeters... 2 2.

### Fluke 805 FC Trillingsmeter

Fluke 805 FC Trillingsmeter Technische gegevens Nu compatibel met Fluke Connect Mobile App De betrouwbare, reproduceerbare, nauwkeurige methode om lagers en algemene trillingsniveaus te controleren. Neem

### SePem. Noise loggers for monitoring water pipe networks sturdy convenient reliable

SePem Noise loggers for monitoring water pipe networks sturdy convenient reliable SePem sturdy convenient reliable Detect water losses early Leaks in water pipe networks can result in significant water

### SPX Model A-360 Azimuth Antenna Rotor Model 1 & 2

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com Model A-360 Azimuth

### Air Volume Instruments

Features and Benefits Model PH731 (Accessories shown on following page) ProHood Capture Hood Model PH731 The PH731 ProHood Capture Hood is a multipurpose electronic air balancing instrument primarily used

### Professional PC correlation SeCorr 300 Perfect results thanks to fully digital technology

Professional PC correlation SeCorr 300 Perfect results thanks to fully digital technology SeCorr 300 The principle SEWERIN has a long tradition of producing systems for locating damages in pipe networks

### ExDetector IRCO2. ExDetector IRHC. Data Sheet. Gas Detection and Warning Systems

ExDetector IRCO2 ExDetector IRHC Gas Detection and Warning Systems Installation / Set-up In combination with a controller system, gas detectors in the ExDetector IR range haven the following functions:

### NIEUW. Extra functionaliteit, sneller testen, en nog steeds zeer robuust

Extra functionaliteit, sneller testen, en nog steeds zeer robuust Met de Fluke 1650-serie multifunctionele testers kunt u testen conform NEN 1010, NEN 3140 en EN61557 NIEUW Geschikt voor Type B aardlekschakelaar

### PIR DC-SWITCH. DC Passive infra-red Detector. Model No. PDS-10 GEBRUIKSAANWIJZING/INSTRUCTION MANUAL

PIR DC-SWITCH DC Passive infra-red Detector Model No. PDS-10 GEBRUIKSAANWIJZING/INSTRUCTION MANUAL Please read this manual before operating your DETECTOR PIR DC-Switch (PDS-10) De PDS-10 is een beweging

### HANDLEIDING - ACTIEVE MOTORKRAAN

M A N U A L HANDLEIDING - ACTIEVE MOTORKRAAN MANUAL - ACTIVE MOTOR VALVE Model E710877 E710878 E710856 E710972 E710973 www.tasseron.nl Inhoud / Content NEDERLANDS Hoofdstuk Pagina NL 1 ALGEMEEN 2 NL 1.1

### I.S.T.C. Intelligent Saving Temperature Controler

MATEN & INFORMATIE I.S.T.C. Intelligent Saving Temperature Controler Deze unieke modulerende zender, als enige ter wereld, verlaagt het energieverbruik aanzienlijk. Het werkt in combinatie met de energy

### Fluke-meetinstrumenten met ATEX-certificering voor explosiegevaarlijke omgevingen

Fluke-meetinstrumenten met ATEX-certificering voor explosiegevaarlijke omgevingen Intrinsiek veilige multimeteren proceskalibrators Conform aan ATEX Intrinsiek veilige Fluke-instrumenten voor metingen

### Groene industrie POWER QUALITY POWER AND SIGNAL QUALITY. Phoenix Contact B.V. 2015-2016

Groene industrie Phoenix Contact B.V. 2015-2016 POWER AND SIGNAL QUALITY POWER QUALITY 24V PWR ERR 24V 0V 0V OUT I GND GND GND ON OFF 1 2 1A 5A NC PACT RCP- Rogowskisp Eerst Power Quality inzichtelijk

### Combined stationary measuring device

Combined stationary measuring device with mobile gas meter Determines the composition and volume of biogas, sewer gas and landfill gas All plant operators producing electricity from biogas, sewer gas or

### WeZoWe - Werkgroep Zonnepanelen Westeinde. Het adres voor al uw installatiewerk!!

WeZoWe - Werkgroep Zonnepanelen Westeinde Het adres voor al uw installatiewerk!! Inhoud: o Over ons o Advies o Montage o Waarom Zonderland o Gerealiseerde projecten o Garantie o Prijzen o Stand o Advertentie

### SPX AX-400 Azimuth & Elevation antenna rotator Model 1 & 2

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com SPX AX-400 Azimuth

### SPL D2 MKII GEBRUIKSAANWIJZING USER MANUAL

SPL D2 MKII GEBRUIKSAANWIJZING USER MANUAL Spl-D2 Next generation Inleiding De SPL5-D2 unit is een geluidsdrukmeter die gekoppeld kan worden aan de SPL5. Het apparaat kan ook als losse geluidsdrukmeter

### Met elke andere Power Quality Analyzer verspilt u slechts energie.

Met elke andere Power Quality Analyzer verspilt u slechts energie. Fluke 430-serie II Power Quality- en energieanalyzers Modellen uit de Fluke 430-serie II Fluke 434, serie II energieanalyzer De Fluke

### Fluke 3000 FC serie meetinstrumenten Technische gegevens

Fluke 3000 FC serie meetinstrumenten Technische gegevens Nu compatibel met Fluke Connect Mobile App Begin vandaag nog met het bouwen van het systeem van meetinstrumenten van morgen, met de digitale multimeter

### AT-7000-EUR geavanceerde kabelzoeker

AT-7000-EUR geavanceerde kabelzoeker Kabels zoeken opnieuw uitgevonden Nauwkeurige resultaten binnen enkele minuten met nieuwe functies en technologieën die het zoeken van kabels en het identificeren van

### Hydrosteel 6000. The Hydrosteel 6000 hydrogen flux monitor is the ultimate portable tool for active sour, HF and high temperature corrosion.

Hydrosteel 6000 Portable, rapid identification of active corrosion & hydrogen damage hotspots. The Hydrosteel 6000 hydrogen flux monitor is the ultimate portable tool for active sour, HF and high temperature

### SPID Azimuth antenna rotator Type: BIG-RAK

SPID Azimuth antenna rotator Type: BIG-RAK Specifications: SPID BIG RAK PWM Azimuth Rotator BIG RAK @ 12V BIG RAK @ 18V Turning torque in-lbs/nm 2,800 in lb/315 Nm 3,600 in lb/403 Nm Brake Torque in lbs/nm

### SPID Azimuth antenna rotator Type: RAK

Gauke Boelensstraat 108 NL-9203 RS Drachten The Netherlands Tel: +31 (0) 512 354 126 GSM: +31 (0) 650 882 889 Fax: +31 (0) 847 187 776 www.rfhamdesign.com E-mail: info@rfhamdesign.com SPID Azimuth antenna

### TiS10, TiS20, TiS40, TiS45, TiS50, TiS55, TiS60, TiS65 Warmtebeeldcamera s

TECHNISCHE GEGEVENS TiS10, TiS20, TiS40, TiS45, TiS50, TiS55, TiS60, TiS65 Warmtebeeldcamera s Built with Compatibel met de Fluke Connect Mobile App De Fluke Performanceserie Gebouwd voor prestaties, met

### liniled Cast Joint liniled Gietmof liniled Castjoint

liniled Cast Joint liniled Gietmof liniled is een hoogwaardige, flexibele LED strip. Deze flexibiliteit zorgt voor een zeer brede toepasbaarheid. liniled kan zowel binnen als buiten in functionele en decoratieve

### Multitec BioControl system BioControl 2. The entry-level model for individual process optimisation

Multitec BioControl system BioControl 2 The entry-level model for individual process optimisation BioControl 2: user-friendly, flexible, efficient The entry-level model for automated process optimisation

### FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE

FOR DUTCH STUDENTS! ENGLISH VERSION NEXT PAGE Tentamen Analyse 8 december 203, duur 3 uur. Voeg aan het antwoord van een opgave altijd het bewijs, de berekening of de argumentatie toe. Als jeeen onderdeel

### Gedetailleerde specificaties

, Ti95 en Ti90 warmtebeeldcamera's voor industriële en Systém Technische gegevens Nu compatibel met Fluke Connect Mobile App SUPERIEURE BEELDKWALITEIT RUIMTELIJKE RESOLUTIE 3,39 mrad Ti 95, Ti90 5,6 mrad

### n Auto current reversal cancels standing emfs n Protected to 600 V n Automatically detects continuity in n Multiple operating modes including fully

DLRO 10 AND DLRO10X DLRO 10 and DLRO 10X n Auto current reversal cancels standing emfs n Protected to 600 V n Automatically detects continuity in potential and current connections n Multiple operating

### Power Quality aspecten van LED-lampen.

Power Quality aspecten van LED-lampen. Hoe meet je die en wat betekent het voor de praktijk? Mark Vloemans AR Benelux Timothy Hertstein ZES ZIMMER Onderwerpen o Aanleiding o Wat is de Power Factor? o Hoe

### HANDLEIDING Honeywell MultiPro multigasmonitor

HANDLEIDING Honeywell MultiPro multigasmonitor INHOUD 1. GEBRUIKTE CONVENTIES... 2 2. AANBEVELINGEN... 2 3. AAN-/UITZETTEN VAN DE MULTIPRO... 4 3.1 Aanzetten van de MultiPro... 4 3.2 Uitzetten van de MultiPro...

### Settings for the C100BRS4 MAC Address Spoofing with cable Internet.

Settings for the C100BRS4 MAC Address Spoofing with cable Internet. General: Please use the latest firmware for the router. The firmware is available on http://www.conceptronic.net! Use Firmware version

### HANDLEIDING DPM TT 570-serie manometer

HANDLEIDING DPM TT 570-serie manometer INHOUD 1. GEBRUIKSBEPERKINGEN... 2 2. BEDIENINGSTOETSEN... 2 3. DISPLAY... 2 4. AKOESTISCH SIGNAAL... 3 5. SIGNAAL IN REFERENTIE... 3 6. RS-232 POORT... 3 7. MEETMODE...

### AQUAPHON. Electro-acoustic water leak detection professional flexible intelligent

AQUAPHON Electro-acoustic water leak detection professional flexible intelligent AQUAPHON professional flexible intelligent Ideal for detecting leaks in water pipe networks When it comes to detecting leaks

### Fluke 810 Vibration Tester

Fluke 810 Vibration Tester Technical Data When you need an answer now The most advanced troubleshooting tool for mechanical maintenance teams who need an answer now. The unique diagnostic technology helps

### Opgave 2 Geef een korte uitleg van elk van de volgende concepten: De Yield-to-Maturity of a coupon bond.

Opgaven in Nederlands. Alle opgaven hebben gelijk gewicht. Opgave 1 Gegeven is een kasstroom x = (x 0, x 1,, x n ). Veronderstel dat de contante waarde van deze kasstroom gegeven wordt door P. De bijbehorende

### Presentation Fire fighter safety switch

Presentation Fire fighter safety switch 20 september 2011 Elsbeth Faasse Cor Jansen Santon Holland BV AGENDA SANTON INTRODUCTION PRODUCTS NEW DEVELOPMENTS The Fire fighter Safety Switch Introductie Diverse

### 1a. We werken het geval voor het tandenpoetsen uit. De concepten zijn (we gebruiken Engelse termen en afkortingen):

Uitwerking Huiswerkopgave Inleiding Modelleren Hoofdstuk 3 1a. We werken het geval voor het tandenpoetsen uit. De concepten zijn (we gebruiken Engelse termen en afkortingen): tube=[cap:{open,close},hand:{l,r,none}]

### HANDLEIDING SenseAir psense (RH) draagbare CO2-meter

HANDLEIDING SenseAir psense (RH) draagbare CO2-meter INHOUD 1. INLEIDING... 2 2. KENMERKEN... 2 3. STROOMVOORZIENING... 2 4. HET INSTRUMENT... 3 4.1. LCD-scherm... 3 4.2. Druktoetsenpaneel... 4 5. BEDIENING...

### TiX560 en TiX520 warmtebeeldcamera's

en warmtebeeldcamera's De Fluke Expert-serie Technische gegevens HOOGWAARDIGE BEELDKWALITEIT RUIMTELIJKE RESOLUTIE en 1,31 mrad RESOLUTIE en 320 x 240 (76.800 pixels) en 640x480 (307.200 pixels) met SuperResolution-modus

### lindab we simplify construction Lindab Lekdichtheidtester Maakt lekdebietmetingen eenvoudiger

lindab we simplify construction Lindab Lekdichtheidtester Maakt lekdebietmetingen eenvoudiger Maakt lekdebietmetingen eenvoudiger Een kompleet toestel voor het ter plaatse meten van luchtlekken in kanalen.

### Enphase M215. De oplossing bij schaduwvorming!

Enphase M215 De oplossing bij schaduwvorming! [ betrouwbaar ] [ productief ] [ slim ] [ veilig ] 1. Enphase micro-omvormer Elke Enphase micro-omvormer is verbonden aan één enkele module en maakt gebruik

### IQ Management System TM

IQ Management System TM IQ Management System TM is a feature-rich portable fleet management software package IQ Management System TM Simplified, automated device tests Instrument identification test Battery

### Product naam: MM03877

Product naam: MM03877 Specificaties MM03877: Levensduur: 15000u Lengte: 103 mm Wattage: 3.50W Diameter: 35 mm Kleur: 2800K (warmwit) Gewicht: 34 gr Lumen: 250 lu Spanning: 230 V RA: 80 Beschermingsklasse:

### HANDLEIDING Honeywell ToxiPro gasmonitor

HANDLEIDING Honeywell ToxiPro gasmonitor INHOUD 1. GEBRUIKTE CONVENTIES... 2 2. AANBEVELINGEN... 2 3. OVERZICHT... 3 3.1 Methodes voor monstername... 4 3.2 Sensoren... 4 3.2.1 Sensor bereik en resoluties...

### ZN-Shine Solar. 300 Wp zonnepaneel. metdezon zonne-energie voor iedereen

metdezon zonne-energie voor iedereen ZN-Shine Solar 300 Wp zonnepaneel Het 300 Wp zonnepaneel van ZN-Shine Solar is een goede keuze. Het paneel levert een hoog rendement en is van uitstekende kwaliteit.

### Druk(verschil)meters. S2600-serie S4600-serie S4600 ST-serie Accessoires

Druk(verschil)meters S2600-serie S4600-serie S4600 ST-serie Accessoires 3 De BLAUWE LIJN druk(verschil)meters zijn onderverdeeld in de S2600-serie, de S4600-serie en de S4600 ST-serie. Deze instrumenten

### OMVORMERS. SolarEdge driefasige omvormers SE4K - SE12.5K. Speciaal ontworpen om te werken met power-optimizers. www.solaredge.com

SE4K - SE12.5K OMVORMERS Speciaal ontworpen om te werken met power-optimizers Superieur rendement (98%) Klein, de lichtste in zijn soort en eenvoudig te installeren Ingebouwde monitoringsysteem op paneelniveau

Handleiding Installatie ADS Versie: 1.0 Versiedatum: 19-03-2014 Inleiding Deze handleiding helpt u met de installatie van Advantage Database Server. Zorg ervoor dat u bij de aanvang van de installatie

### Pesten onder Leerlingen met Autisme Spectrum Stoornissen op de Middelbare School: de Participantrollen en het Verband met de Theory of Mind.

Pesten onder Leerlingen met Autisme Spectrum Stoornissen op de Middelbare School: de Participantrollen en het Verband met de Theory of Mind. Bullying among Students with Autism Spectrum Disorders in Secondary

### HANDLEIDING BLAUWE LIJN S2600-serie druk(verschil)meters

HANDLEIDING BLAUWE LIJN S2600-serie druk(verschil)meters S2601 S2680 Druk aansluitingen LCD met displayverlichting Beschermholster met geïntegreerde magneet Escapetoets Entertoets Holdtoets Keuzetoets

### SHP-TS TwinArc SA SHP-TS 400W TWINARC E40 SL PRODUCT OVERVIEW

Range Features Range of high pressure sodium lamps with double arc tube construction Dual arc tube design guarantees immediate re-strike after a power interruption Doubled lamp life and reduced occurrence

### Gebruik van het LOGO in geautomatiseerde verkiezingen

BIJLAGE 1 S.A. STERIA Benelux N.V. Gebruik van het LOGO in geautomatiseerde verkiezingen Technische bepalingen voor de weergave van het logo op de schermen. Versie 1.2 Guy JASPERS Revisions Revision Description

### 1148-CPR-20080603. SAFETY-PRODUCT nv Beverlosesteenweg 100 B-3580 BERINGEN EN 40-5:2002

Keizerinlaan 66 B-1000 BRUSSEL Certificaat van prestatiebestendigheid 1148-CPR-20080603 Overeenkomstig de Verordening 305/2011/EU van het Europees Parlement en de Raad van 9 maart 2011 (de Bouwproductenverordening

### FHI dag 3 april 2013 Wiep Folkerts, directeur SEAC

FHI dag 3 april 2013 Wiep Folkerts, directeur SEAC Innovaties in Zonne-energie ~~~~ een bloeiend stukje Nederlandse kenniseconomie Waarom zonne-energie? Prijsdaling PV modules Average module spot price

### CHROMA STANDAARDREEKS

CHROMA STANDAARDREEKS Chroma-onderzoeken Een chroma geeft een beeld over de kwaliteit van bijvoorbeeld een bodem of compost. Een chroma bestaat uit 4 zones. Uit elke zone is een bepaald kwaliteitsaspect

### Ti450, Ti400 and Ti300 warmtebeeldcamera's De Fluke Professional-serie

TECHNISCHE GEGEVENS Ti450, Ti400 and Ti300 warmtebeeldcamera's De Fluke Professional-serie Built with Geschikt voor Fluke Connect SUPERIEURE BEELDKWALITEIT RUIMTELIJKE RESOLUTIE Ti450 en Ti400 1,31 mrad

### PROFITEST. E-Installatietester MASTER TESTER INSTALLATION

PROFITEST INSTALLATION MASTER TESTER 1 E-Installatietester Praktisch design Met het nieuwe meetapparaat PROFITEST MASTER kunnen elektroprofessionals beschikken over een universeel meetinstrument van het

### COGNITIEVE DISSONANTIE EN ROKERS COGNITIVE DISSONANCE AND SMOKERS

COGNITIEVE DISSONANTIE EN ROKERS Gezondheidsgedrag als compensatie voor de schadelijke gevolgen van roken COGNITIVE DISSONANCE AND SMOKERS Health behaviour as compensation for the harmful effects of smoking

### Johto. Flexible light

Johto Flexible light Johto is a high quality lighting system based on LED for technically sophisticated interior and exterior light. It provides homogeneous and dot free illumination in very low installation

### WWW.EMINENT-ONLINE.COM

WWW.EMINENT-OINE.COM HNDLEIDING USERS MNUL EM1016 HNDLEIDING EM1016 USB NR SERIEEL CONVERTER INHOUDSOPGVE: PGIN 1.0 Introductie.... 2 1.1 Functies en kenmerken.... 2 1.2 Inhoud van de verpakking.... 2

CUSTOMIZE YOUR APPLE AND PRESENT A UNIQUE COMPANY GIFT Ready for customization according to your companies wishes? The Royal Blue Collection a Custom Made collection. This is the perfect combination of

### Introduction Henk Schwietert

Introduction Henk Schwietert Evalan develops, markets and sells services that use remote monitoring and telemetry solutions. Our Company Evalan develops hard- and software to support these services: mobile

### HANDLEIDING Airflow LoFlo balometer

HANDLEIDING Airflow LoFlo balometer INHOUD 1. SPECIFICATIES...2 2. ALGEMENE BESCHRIJVING...2 3. BEDIENINGSINSTRUCTIES...2 4. VEILIGHEID...3 5. INSTRUMENT SET-UP EN OPSLAG...3 6. OVER DE DISPLAY VAN DE

### Uitwegen voor de moeilijke situatie van NL (industriële) WKK

Uitwegen voor de moeilijke situatie van NL (industriële) WKK Kees den Blanken Cogen Nederland Driebergen, Dinsdag 3 juni 2014 Kees.denblanken@cogen.nl Renewables genereren alle stroom (in Nederland in

### Ontpopping. ORGACOM Thuis in het Museum

Ontpopping Veel deelnemende bezoekers zijn dit jaar nog maar één keer in het Van Abbemuseum geweest. De vragenlijst van deze mensen hangt Orgacom in een honingraatpatroon. Bezoekers die vaker komen worden

### Bijlage 2: Informatie met betrekking tot goede praktijkvoorbeelden in Londen, het Verenigd Koninkrijk en Queensland

Bijlage 2: Informatie met betrekking tot goede praktijkvoorbeelden in Londen, het Verenigd Koninkrijk en Queensland 1. Londen In Londen kunnen gebruikers van een scootmobiel contact opnemen met een dienst

### Rookgasmeter Instrument afgebeeld schaal 1:1

BLUE YZER ST Rookgasmeter Instrument afgebeeld schaal 1:1 Veel moderne cv-installaties worden voor optimale werking en veiligheid afgesteld op CO 2 (kooldioxide). Hiervoor is meting van de samenstelling

### MIT515, MIT525, MIT1025, MIT1525

MIT515, MIT525, MIT1025, MIT515, MIT525, MIT1025, Meet tot 35 TΩ Veiligheidsclassificatie tot CAT IV 1000 V tot 3000 m Uniek dubbel kastontwerp - extra bescherming voor de gebruiker Werkt op batterij of

Version 1.2 DECLARATION FOR GAD approval Declare that for the following central heating boilers Intergas Calderas de Calefacción S. L. Kombi Kompakt R 24, 28/24, 36/30 and Prestige The installation and

### Workflow en screenshots Status4Sure

Workflow en screenshots Status4Sure Inleiding Het Status4Sure systeem is een ICT oplossing waarmee de transportopdrachten papierloos door het gehele proces gaan. De status kan gevolgd worden door de logistieke

### Tentamen Thermodynamica

Tentamen Thermodynamica 4B420 4B421 10 november 2008, 14.00 17.00 uur Dit tentamen bestaat uit 4 opeenvolgend genummerde opgaven. Indien er voor de beantwoording van een bepaalde opgave een tabel nodig